Remove this adapter Agilent EA before making measurements. The same limit fail field is used by both channel and window limits. Use the , and keys to move between the name, frequency, calibration factors and offsets. Repeatedly pressing accesses a menu which allows you to increase and decrease the display contrast. In this instance, channel A averaging is calculated with a resolution of four and channel B with a resolution of two. Page 74 If you measure a signal with a frequency outside the frequency range defined in the sensor calibration table, the power meter uses the highest or lowest frequency point in the sensor calibration table to calculate the calibration factor. Option CFT but can be overridden dependent offset by this softkey.
|Date Added:||9 March 2009|
|File Size:||8.57 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
E4419B EPM Series Dual-Channel Power Meter
Personal Safety Considerations Any adjustments or service procedures that require operation of the C A U T I O N instrument with protective covers removed should be performed only by trained service personnel. Page that is, when this key is off but the power is connected to the instrument the red LED is lit. Wait Symbol This pop-up window is displayed when you are required to modify numeric, or alphanumeric data.
If you measure a signal with a frequency outside the frequency range defined in the sensor calibration table, the power meter uses the highest or lowest frequency point in the sensor aglent table to calculate the calibration factor.
HP Agilent Keysight, EB
To obtain a carry case, order Agilent part number A. Page 80 for further information.
This label has two instrument identification entries. Setting Averaging A – channel B and the lower window has a resolution of four and is measuring channel A.
Replacing the battery is basically a reversal of the removal procedure. These errors also display the error annunciator and are written to the error queue.
Notice that the lower window is now set up to e4419b a channel A over channel B ratio measurement.
This can either be a direct channel measurement, a ratio e419b of both channels or a difference measurement between both channels. A series of bitmaps are displayed on the power meter showing: Page 30 85 for further information. TTL inputs are also provided to initiate zero and calibration cycles.
Keysight Technologies (Agilent HP) EB used or new for sale at used-line
Chapters Table Of Contents A pop-up window displays the parameter. At least an 18AWG earthing conductor should be used in such an instance, to ground the instrument to an assured earth terminal. If this occurs, contact your Agilent Technologies Sales and Service office for instructions on returning the power meter to Agilent Technologies for service. Page A discrete parameter was received which was not a valid choice for the command.
The same limit fail field is used by both channel and window limits.
Test Descriptions Four tests are available for the serial interface: To select the interface language from the front panel: Error Messages — Exponent too large A numeric parameter was found whose exponent was larger than 32, To rack mount two instruments side by side, order Option or lock-link kit and flange kit You can search the DoC by its product model or description. Press this softkey to modify the highlighted parameter, which is either a ee4419b, calibration factor, offset or table name.
Step Detection ChA Filter softkey to access the filter menu. Enter text from picture: If both TTL inputs are simultaneously low under any circumstances other than those shown above, the operation is undefined.
Zero Drift of Sensors Power sensor dependent refer to Table For Agilent E- Series power sensors e441b normal and x2 speed modes, manual filter, 10 dB decreasing power step: Single channel measurements can be displayed relative to a stored reference. This opens a position at the end of the queue for a new error message, if one is subsequently detected.
Calibrating The Power Meter During calibration, the power meter automatically switches the power reference calibrator N O T Wgilent on if it is not already onthen after calibration, it switches it to the state it was in prior to the calibration.
Four tests are available for the serial interface: Don’t show me this message again.